Hanna HI83414 Benchtop Turbidity and Chlorine Meter
HI 83414 features four measuring ranges: ratio, non ratio, free chlorine, total chlorine
Meets USEPA requirements
Exclusive chlorine CAL CHECK™ calibration validation
Two, three, four or five point turbidity calibration
USB PC connectivity
On-screen tutorial and help modes
Log and recall up to 200 measurements
Contextual help and tutorial mode
HI 83414 is a highly accurate dual parameter instrument that reflects HANNA’s years of experience. The HI 83414 successfully combines turbidity and colorimetric measurements to test the most important parameters of drinking water: turbidity and free/total chlorine. The meter is specially designed for water quality measurements, providing reliable and accurate readings on low turbidity and chlorine values. The HI 83414 meets and exceeds the requirements of USEPA and Standard Methods both for turbidity and colorimetric measurements.
With the powerful CAL CHECK™ function, reliable performance can be validated at any moment by using the exclusive HANNA ready-made, NIST traceable standards. A one-point calibration can be performed using the same CAL CHECK™ standard.
This instrument features an optical system to guarantee accurate results, assure long term stability and minimize stray light and color interferences. They also compensate for variations in intensity of the lamp for less frequent calibration.
The 525 nm interference filter of the colorimeter assures accurate and repeatable results. Repeatability of the measurements are ensured with 25 mm round cuvettes made from special optical glass.
Turbidity measurements can be made in the 0.00 to 4000 NTU (Nephelometric Turbidity Units) range when ratiometric measurements are used and in the 0.00 to 40.0 NTU range when non ratio method is used. The HI?83414 has an EPA compliance reading mode which rounds the reading to meet EPA reporting requirements. Alternative EBC and Nephelos measuring units are available. Depending on the measured sample and needed accuracy, normal measurement, continuous measurement or signal averaging measurement can be selected.
A two, three, four or five-point calibration can be performed by using the supplied (<0.1, 15, 100, 750 and 2000 NTU) standards. If user prepared standards are used, the calibration points can be modified. Free or total chlorine measurements can be made in the 0.00 to 5.00 mg/L (ppm) range.
HI 83414 features complete GLP (Good Laboratory Practice) functions that allow traceability of the calibration conditions. The last calibration points, time and date can be checked.
This meter also incorporates a user-friendly interface with an easy to understand, graphic LCD. All messages are in plain text making them easy to read. Comprehensive contextual help is available at the press of a button. All messages and help screens are available in several languages. Confirmation and error acoustic signals help the user during instrument operation. Furthermore, a tutorial mode of operation guides the user step by step through the analysis process.
The logging function offers complete measurement information. Up to 200 measurements can be stored in the internal memory and consulted at any time. For further storage or analysis options, data can be downloaded to a PC using the USB port.
Order Information: HI 83414-02 (230V) is supplied with sample cuvettes and caps (5), calibration cuvettes for turbidimeter and colorimeter , silicone oil (HI 98703-58), cloth for wiping cuvettes, scissors, power cord and instruction manual.
Range Non Ratio Mode 0.00 to 9.99; 10.0 to 40.0 NTU; 0.0 to 99.9; 100 to 268 Nephelos 0.00 to 9.80 EBC
Ratio Mode 0.00 to 9.99; 10.0 to 99.9; 100 to 4000 NTU 0.0 to 99.9; 100 to 26800 Nephelos 0.00 to 9.99; 10.0 to 99.9; 100 to 980 EBC
Free and Total Cl 0.00 to 5.00 mg/L (ppm)
Resolution Non Ratio Mode 0.01; 0.1 NTU; 0.1; 1 Nephelos; 0.01 EBC
Ratio Mode 0.01; 0.1; 1 NTU; 0.1; 1 Nephelos; 0.01; 0.1, 1 EBC
Free and Total Cl 0.01 mg/L (ppm) from 0.00 to 3.50 mg/L (ppm) 0.10 above 3.50 mg/L (ppm)
Accuracy @25°C Turbidity ±2% of reading plus 0.02 NTU (0.15 Nephelos; 0.01 EBC) ±5% of reading above 1000 NTU (6700 Nephelos; 245 EBC)
Free and Total Cl ±0.03 mg/L or ±3% of reading (whichever is greater)
Method Turbidity Nephelometric method (90°) or Ratio Nephelometric Method (90° & 180°), Adaptation of the USEPA Method 108.1 and Standard Method 2130 B
Free and Total Cl adaptation of the USEPA Method 330.5 and Standard Method 4500-Cl G.
Light Detector Turbidity silicon photocell
Free and Total Cl silicon photocell with 525 nm narrow band interference filters
Standards Turbidity <0.1, 15, 100, 750 and 2000 NTU
Free and Total Cl 1.00 mg/L free/total chlorine
Calibration Turbidity two, three, four or five-point calibration
Free and Total Cl one-point calibration
Turbidity Range Selection
Jual Hanna HI83414 Benchtop Turbidity and Chlorine Meter harga kompetitif Tentunya Gratis antar untuk Area Jakarta dan dapat dikirim keseluruh Indonesia, Jika Anda butuh bantuan silahkan hubungi Customer Service (026) 5540 5390 & Whatsapp di 0813-1864-0412 dan Email firstname.lastname@example.org