MITECH MXT-1600 Portable Belt Flaw Detector
Mitech MXT series portable belt flaw detector, based on short wave radiography principle, use flat-type image intensifier, built-in high-voltage generator and SD memory card to achieve a convenient belt non-destructive testing and data storage records. Its performance is stable, high sensitivity, clear imaging and easy operation, with real-time imaging, camera storage and other functions. Effective for coal mining enterprises mining equipment (belt) testing, it is a professional non-destructive testing equipment to save production costs, eliminate the necessary hidden dangers.
Function & application
- Coal mining enterprises mining equipment (belt) detection.
- Experimental teaching experiment in colleges and universities.
- Scientific research institutions of material analysis testing.
- Wire rope belt manufacturing enterprises quality inspection links.
The portable belt flaw detector is a new generation of short-wave radiation detection equipment in Mitech.Which is generated by the built-in high-voltage emission device short-wave radiation (spectrally similar to the natural X-ray),through the collimator to limit the emission beam, penetrate the measured belt into the short-wave radiation sensitive flat-type image intensifier,The use of different density substances on the short-wave radiation absorption rate is different from the steel cord and the belt inside and outside the defects of abnormal imaging on the image intensifier screen, and then converted into a visible view of the belt to achieve the nondestructive testing of the belt.
- Operating temperature：10℃～45℃；
- Relative humidity：≤75％；
- Atmospheric pressure range：700 hPa～1060hPa
- No corrosive media around, no strong magnetic field interference;
- Can test wire rope in the belt twitch,broken rope Rust rope .Widely used in coal mining companies such as belt inspection, quality inspection department quality inspection, institutions of higher learning, scientific research test, etc;
- Don’t need a dark room and protective, it can realize direct observation real time, and improve testing efficiency conveniently and quickly;
- Using flat image intensifier, with high sensitivity,it responses quickly, can display real-time imaging;
- With self-protection function, it can auto-power off after working 60s to avoid high temperature damage the instrument;
- With 2 rechargeable batteries, it can get rid of the 220V power supply and meet the long working time demand
- With integration design, small volume, light weight, solid in construction, it can be carried by hand, low radiation dose, safe and reliable;
- Built-in SD card , it can take real-time photos stored images, for subsequent processing analysis;
- A software system connected with PC can realize real-time video function.
- Meet domestic and foreign standards.
- Please insert batteries or connect the power supply under close inspection work patterns, to avoid damaging the instrument;
- Before use, the separation of the protective plate should be installed,When operating, the hand is strictly prohibited than the shield screen extension line;
- Please avoid continuous testing for a long time, usually 1 minute to stop testing for three minutes rest, and shut down after the shutdown time to 5 seconds or more, to avoid high pressure accumulation, resulting in man-made equipment failure;
- When using,if you find crescent-shaped defects in the image , there are oil in the aluminum box or other abnormalities, you should immediately stop using, and contact the manufacturer;
- After use,please put the instrument in the aluminum box to reserve;
- When using and storage,should pay attention to waterproof, so as to avoid circuit failure, affecting the normal use;
- Test the instrument for periodic leak detection and if found that excessive leakage of radiation should stop using;
- The maintenance of the tube or image intensifier are required to return to the factory, if user does self-maintenance, the company will not be responsible for the safety of instrument radiation;
- The instrument should be regularly cleaned and maintained, can be dipped in a small amount of alcohol wipe;
- Please do not disassemble the instrument, contact the company on a regular basis to maintain the service (about two years ) to ensure safety, please contact the Department of Mitech Instrument Service, 4000600280.
Specifications MITECH MXT-1600 Portable Belt Flaw Detector
|Grey level||≥7 level|
|Maximum width (Can be customized)||800mm|
|Short-wave ray tube voltage||75kV|
|Short-wave ray tube current||0.4mA|
|Focus nominal value||0.1mm|
|Power supply||DC14.4V or AC220V 50Hz|
|Charging power supply||AC220V 50Hz|
Produk asli Kami distributor langsung Mitech . Harga murah dan berkualitas bergaransi tentunya. dapat di kirim ke seluruh wilaya indonesia.
Dalam pengiriman produk yang pelanggan beli, sebelumnya sudah kami cek untuk memastikan produk dalam keadaan baik dan siap kirim.